In the petroleum industry, cloud point refers to the temperature below which wax in diesel or biowax in biodiesels forms a cloudy appearance. The presence of solidified waxes thickens the oil and clogs fuel filters and injectors in engines. The wax also accumulates on cold surfaces (producing, for example, pipeline or heat exchanger fouling) and forms an emulsion with water. Therefore, cloud point indicates the tendency of the oil to plug filters or small orifices at cold operating temperatures.
An everyday example of cloud point can be seen in olive oil stored in cold weather. Olive oil begins to solidify at around 4 °C, whereas winter temperatures in temperate countries can often be colder than 0 °C. In these conditions, olive oil begins to develop white, waxy clumps of solidified oil that sink to the bottom of the container.
In crude or heavy oils, cloud point is synonymous with wax appearance temperature (WAT) and wax precipitation temperature (WPT).
The cloud point of a nonionic surfactant or glycol solution is the temperature at which the mixture starts to phase-separate, and two phases appear, thus becoming cloudy. This behavior is characteristic of non-ionic surfactants containing polyoxyethylene chains, which exhibit reverse solubility versus temperature behavior in water and therefore \”cloud out\” at some point as the temperature is raised. Glycols demonstrating this behavior are known as \”cloud-point glycols\” and are used as shale inhibitors. The cloud point is affected by salinity, being generally lower in more saline fluids.
Automatic Measuring Method
ASTM D5773, Standard Test Method of Cloud Point of Petroleum Products (Constant Cooling Rate Method) is an alternative to the manual test procedure. It uses automatic apparatus and has been found to be equivalent to test method D2500.
The D5773 test method determines the cloud point in a shorter period of time than manual method D2500. Less operator time is required to run the test using this automatic method. Additionally, no external chiller bath or refrigeration unit is needed. D5773 is capable of determining cloud point within a temperature range of -60 °C to +49 °C. Results are reported with a temperature resolution of 0.1 °C.
Under ASTM D5773, the test sample is cooled by a Peltier device at a constant rate of 1.5 +/- 0.1 °C/min. During this period, the sample is continuously illuminated by a light source. An array of optical detectors continuously monitor the sample for the first appearance of a cloud of wax crystals. The temperature at which the first appearance of wax crystals is detected in the sample is determined to be the cloud point.